- Research
- Open Access
- Published:

# Simultaneous suppression of disturbing fields and localization of magnetic markers by means of multipole expansion

*BioMagnetic Research and Technology*
**volumeÂ 2**, ArticleÂ number:Â 6 (2004)

## Abstract

### Background

Magnetically marked capsules serve for the analysis of peristalsis and throughput times within the intestinal tract. Moreover, they can be used for the targeted disposal of drugs. The capsules get localized in time by field measurements with a superconducting quantum interference device (SQUID) magnetometer array. Here it is important to ensure an online localization with high speed and high suppression of disturbing fields. In this article we use multipole expansions for the simultaneous localization and suppression of disturbing fields.

### Methods

We expand the measurement data in terms of inner and outer multipoles. Thereby we obtain directly a separation of marker field and outer disturbing fields. From the inner dipoles and quadrupoles we compute the magnetization and position of the capsule. The outer multipoles get eliminated.

### Results

The localization goodness has been analyzed depending on the order of the multipoles used and depending on the systems noise level. We found upper limits of the noise level for the usage of certain multipole moments. Given a signal to noise ratio of 40 and utilizing inner dipoles and quadrupoles and outer dipoles, the method enables an accuracy of 5 mm with a speed of 10 localizations per second.

### Conclusion

The multipole localization is an effective method and is capable of online-tracking magnetic markers.

## Background

The transport of capsules in the alimentary tract underlies complex influencing factors like the patients peristalsis, the hydration and the form and size of the capsules. A procedure which allows the instantaneous localization of the capsules supports a number of patient examinations as well as examinations of new drug forms [1â€“4]. Capsules can be marked radioactively (scintigraphy) or magnetically. The scintigraphy [5] has a lower time resolution compared to the magnetic localization, and due to radiation it is not appropriate for examinations with healthy probands.

The localization of magnetically marked capsules (magnetic markers) must be spatially accurate and with high temporal resolution. For the spatial localization the marker field must be separated from the external magnetic disturbing fields. This separation can be achieved by splitting the magnetic field in multipole moments [6]. The method proposed utilizes the multipole moments directly for the determination of the position and the magnetic moment of the marker. Thus, the separation of disturbing fields and the localization are integrated numerically effective into one procedure. This allows a fast online-localization of the marker capsules.

Multipole expansions are used also to model spatially distributed biological sources such as brain currents [7, 8].

The application of multipoles for the localization of magnetic dipoles is described in [9, 10], and is used in other technical areas without disturbing field suppression [11].

Marking of capsules and pills takes place by partially filling them with black iron oxide (Fe_{3}O_{4}) which is subsequently magnetized up to saturation. The magnetic field measurement is performed within magnetically shielded rooms by the use of highly sensitive SQUID arrays. For the investigation at hand we conduct simulation runs to determine the performance of the multipole localization.

## Methods

### Algorithm

The field of a magnetic marker located adjacent to the point of origin can be expressed by a multipole expansion in Cartesian coordinates (*x*_{1}, *x*_{2}, *x*_{3}). If the distance between marker and origin is small compared to the distance between a magnetic sensor and the origin, the field of the marker at the sensor position is given by the first elements of the multipole expansion. With the notation

follows

with *c*^{m} being the dipole, quadrupole and octopole moments of the field expansion.

The form functions

arise from a Taylor series expansion in the parameter with and . It holds

With the Kronecker delta

follows

and

Conversely, a Taylor series expansion of â€“ compared to the sensor coordinates â€“ far away located field sources in the parameter

with yields a multipole expansion of external disturbing fields:

We denote the multipole moments *c*^{ex} of the expansion of fields of external sources as "outer moments" to distinguish them from the "inner moments" *c*^{m}.

To get the same normalization and symmetry properties for the outer and inner form functions, we define the outer form functions

by

It follows

and

The tensors of 3^{rd} order and of 4^{th} order own the following symmetry features which are identical for the inner and outer multipole expansion:

We combine the resulting 3, 5 and 7 linearly independent components of the tensors of 2^{nd}, 3^{rd} and 4^{th} order to one vector for the marker field and one vector for the external disturbing field :

The summation of equation (1) and equation (6) yields the field expansion for a magnetic marker with disturbing fields. We truncate this expansion after the octopole terms, and transcript it into a linear equation system for the determination of equivalent multipole moments **c** for a measurement **B**_{meas}:

The structure of the vectors **B**_{meas} and **c** and the matrix **F** is given below in the formulas (15...24). The residuals 0(Â·) are sufficiently small, if the coordinates of the marker are small compared to the coordinates of the field sensors , and if the coordinates of the field sensors are small compared to the coordinates of the external disturbing field sources .

**B**_{meas} is a vector with the measurement values of the magnetometer sensor field in the positions with the directions :

The matrix **F** is built from the linearly independent form functions for inner and outer field sources given in equation (13). Their scalar product with the sensor normal directions yields one row for every sensor:

The number of columns of **F** is the sum of the numbers of inner and outer field functions used. Each column describes the field of one specific magnetic moment with unit strength measured by the sensor system. The Matrix **F** is called the forward matrix of all moments considered. The Matrix **F** is structured into submatrices for different moments:

Matrix

and are the forward matrices for inner and outer dipoles:

Matrix

and are the forward matrices for inner and outer quadrupoles. The size of is (*Nsen*/5) with the rows belonging to quadrupole moments with indices (1,1; 3,3; 1,2; 2,3; 3,1).

Matrix

and are the forward matrices for inner and outer octopoles. The size of is (*Nsen*/7) with the rows belonging to octopole moments with indices (1,2,2; 2,3,3; 3,1,1; 1,3,3; 2,1,1; 3,2,2; 1,2,3).

The vector of multipole moments **c** is composed of inner and outer dipole moments **c**_{d}, quadrupole moments **c**_{q} and octopole moments **c**_{o}:

The inner dipole moments

describe a dipole at the point of origin, the outer dipole moments describe a homogeneous disturbing field:

The

represent a quadrupole at the point of origin. The describe an external gradient field, whose field strength vanishes at the origin and which has no spatial derivations of 2^{nd} or higher order. This field can be measured by five ideal gradiometers at the origin and can be compared with the creation of software gradiometers.

The

represent an octopole at the origin. The describe an external gradient field of 2^{nd} order, which has no spatial derivations of 3^{rd} or higher order and whose field strength and spatial derivatives of 1^{st} order vanish at the origin. This field could be measured by 7 ideal second order gradiometers at the origin, it can be compared with the creation of software gradiometers of 2^{nd} order:

Due to its small spatial extension, the magnetic marker can be described as a dipole of strength

at position as a good approximation. The field of this dipole is

With the Taylor series expansion

follows in analogy to equation (1)

A comparison of coefficients of (1) and (27) yields

and

The dipole strength

can be determined by the dipole moment . An equation system for the adjacent calculation of the dipole position from the dipole strength and the quadrupole moment follows from (29) and (23):

with

This equation system is named shift equation in analogy to [10]. It is overdetermined, and can be solved by means of the pseudo inverse of **m**.

We get the multipole moments **c**, which are required for the localization of the marker dipole, from solving the overdetermined equation system (14) by means of the pseudo inverse of **F**:

**c** = (**F**^{T}Â·**F**)^{-1}Â·**F**^{T}Â·**B**_{meas}. Â Â Â (33)

Here, the matrix of form functions **F** must contain columns at least for the moments and .

Iterative dipole localization for a fixed dipole (e.g. one time point) is achieved by using the localization position as a new point of origin. The step length of the last localization step serves as a stop criterion for the iterative localization procedure. This is justified by considering the residuals of equation (14) within the convergence range of the procedure, and will practically be shown by the results of the following simulations.

The tracking of a moved dipole based on measurements at consecutive time steps works by updating both the point of origin and the measurement data set after each localization step (Fig. 1). The localization step must be monitored, since it contains information about the marker speed and the noise dependent and speed dependent localization errors.

### Measurement system

The simulations to determine the performance of the algorithm use the sensor geometry of the multi channel SQUID system Argos 200 from AtB (Advanced Technologies Biomagnetics, Pescara, Italy). The ARGOS 200 system contains fully integrated planar SQUID magnetometers produced using Nb technology with integrated pick-up loops. The sensing area is a square of 8 mm side length. The intrinsic noise level of the built in 195 SQUID sensors is below 5 fT Hz^{-1/2} at 10 Hz. Three sensors form one orthogonal triplet in each case. The measurement plane with a diameter of 23 cm consists of 56 of those triplets. The reference array consists of seven SQUID sensor triplets located in the second level in a plane which is positioned parallel to the measurement plane at a distance of 98 mm. The third (196 mm above the first plane) and fourth (254 mm above the first plane) levels contain one triplet each (Fig. 2).

The measurement system is positioned within a magnetically shielded room, consisting of 3 highly permeable shieldings and one eddy current shielding. The shielding performance is 38 db at 1 Hz, 55 db at Hz and 80 db at 20 Hz.

The sensor arrangement in orthogonal triplets facilitates the measurement of all 3 spatial components of the magnetic field. Thus, the required field coverage for the localization of a magnetic marker with unknown dipole strength is achieved.

The subdivision into 168 measurement and 27 reference sensors is meant for the creation of software gradiometers. We can use all sensors simultaneously for the multipole method which integrates the suppression of disturbing fields.

With the above described measurement system we performed simulations with different signal to noise ratios.

## Results

We examined the localization characteristics of the multipole method by means of simulation runs at the sensor geometry of the measurement system Argos 200 (Fig. 2). All simulations performed are based on a dipole at position (*x*, *y*, *z*) = (0, 0, -300 mm), i.e. 30 cm below the measurement plane, with a dipole strength of 20 Amm^{2}. This is a realistic dipole position for an examination within the digestive tract. The dipole field was superimposed by uncorrelated, Gauss distributed noise. The noise level in fT is also given as signal-to-noise-ratio (SNR), based on the channel with the strongest amplitude of the dipole field.

The average localization accuracy over 100 simulations has been determined depending on the noise level and on the number of the multipole moments used in the vector **c** (21) (Fig. 3).

The localization was run up to a stable point. We define the localization error as the mean quadratic error of the 100 stable points based on the true dipole position. The localization error increases if we use higher order multipole moments. This holds true for the inner moments **c**^{m} and for the outer moments **c**^{ex} as well. As a good approximation the interrelationship between noise level and localization error is linear, with raising proportionality factor for higher mode numbers. This corresponds to parallel translation of the curves in double logarithmic plotting.

We examined the localization speed depending on the distance of the starting point to the dipole position. For any tested distance the starting point has been moved from the dipole position into 100 random directions. The remaining mean distance to the dipole position after one localization step is depicted in Fig. 4. The localization speed turns to be significantly higher when using inner octopoles. It gets higher with a shorter starting distance in both an absolute and a relative manner based on the starting distance. Both effects are to be expected directly from the residuals of equation (14). The influence of the outer multipoles on the localization speed is low. The convergence radius at which the dipole was found from all 100 directions decreases slightly with the raising number of outer multipoles used, and increases slightly if inner octopoles are used (unequal right ends of the respective curves in Fig. 4). The convergence radius ranges between 6 and 10 cm. The maximum number of iterations for a target accuracy of 1 mm can be estimated from Fig. 4 as 3.

In the following we examined the interrelationship between the convergence distance at y-direction and the noise level. The maximum y-distance of the starting position to the dipole, at which the dipole could be found with 100 random noise distributions, is depicted in Fig. 5. It shows that the convergence distance remains unchanged almost up to the point of critical noise level (see Fig. 3) at which localization becomes impossible. The convergence distance, also compare the maximum convergence radius from the curve ends of plot (Fig. 4), depends only marginally on the choice of the inner ansatz functions. It decreases slightly when using outer multipoles. Having a convergence radius of at least 6 cm for the dipole position tested, the choice of the starting position can be regarded as noncritical.

The computing time used for one localization step is 5 ms with an implementation in Matlab at a standard Windows PC with a 2 GHz clock frequency. With maximum 3 iterations per localization step and additional computing time needed for data transfer and a basic visualization, 10 localizations per second are possible. This rate is normally sufficient for marker localizations.

## Discussion

The localization speed rises when using inner octopoles

, but this is associated with a higher localization error. At a signal to noise ratio lower than 10^{3} inner octopoles cannot be used. An SNR of at least 10^{2} is required for a source positioned 30 cm below the measurement plane.

The outer moments **c**^{ex} used enlarge the localization error depending on the uncorrelated sensor noise, as shown in Fig. 3. Contrary, the localization error depending on the spatially correlated residual field within the measurement room lowers when using outer moments. Depending on the ratio between correlated and uncorrelated noise which has to be found with practical test series, noise suppression of homogeneous disturbing fields using and possibly noise suppression of gradient fields using are applicable. To ensure convergence, the starting point for the algorithm has to be within the convergence radius given in Fig. 4. With typical measurement conditions, thus, a starting point 10 cm below the center of the measurement system will suffice.

## Conclusions

The multipole localization is an effective algorithm because it unites a method for the suppression of disturbing fields with a localization method. It can be used iteratively and online for the tracking of magnetic marker timelines within the intestinal tract.

## References

Weitschies W, Hartmann V, Grutzmann R, Breitkreutz J: Determination of the disintegration behavior of magnetically marked tablets. Eur J Pharm Biopharm. 2001, 52: 221-226. 10.1016/S0939-6411(01)00178-3.

Kosch O, Osmanoglou E, Weitschies W, Wiedenmann B, MÃ¶nnikes H, Trahms L: Online localisation of a magnetized capsule for investigation of the gastrointestinal passage. Biomed Tech. 2003, 48 (Suppl 1): 244-245.

MÃ¶nnikes H, Osmanoglou E, Kosch O, Fach K, van der Voort IR, Strenzke A, Weitschies W, Wiedenmann B, Trahms L: Evaluation of esophageal motility and transport of solid drug dosage forms by manometry and magnetometry. Biomed Tech. 2003, 48 (Suppl 1): 376-377.

Kosch O, Osmanoglou E, Hartmann V, Strenzke A, Weitschies W, Wiedenmann B, MÃ¶nnikes H, Trahms L: Investigation of gastrointestinal transport by magnetic marker localization. Biomed Tech (Berl). 2002, 47 (Suppl 1 pt 2): 506-509.

Mughal MM, Marples M, Bancewicz J: Scintigraphic assessment of esophageal motility: what does it show and how reliable is it?. Gut. 1986, 27: 946-953.

Taulu S, Kajola M, Simola J: The Signal Space Separation Method. Abstract book of NFSI. 2003, Chieti, Italy, A79-

Nolte G, Curio G: Current multipole expansion to estimate lateral extent of neuronal activity: a theoretical analysis. IEEE Trans Biomed Eng. 2000, 47: 1347-55. 10.1109/10.871408.

Nolte G, Curio G: On the calculation of magnetic fields based on multiple modeling of focal biological current sources. Biophys J. 1997, 73: 1253-62.

Wikswo JP, Swinney KR: A comparison of scalar multipole expansions. J Appl Phys. 1984, 56: 3039-49. 10.1063/1.333885.

Wikswo JP, Swinney KR: Scalar multipole expansions and their dipole equivalents. J Appl Phys. 1985, 57: 4301-08. 10.1063/1.334589.

Gilbert HC, Saglembeni A, Rakijas M, Kohnen KK: Linear perturbation method for kalman filter tracking of magnetic field sources. United States Patent No.: US 6.292.758 B1. Sep 18, 2001

## Acknowledgements

This work has been partially supported by the joint research project "MagMon/NET0158" within the "InnoNet" program of the German Federal Government Department of Research (BMWA).

## Author information

### Authors and Affiliations

### Corresponding author

## Authorsâ€™ original submitted files for images

Below are the links to the authorsâ€™ original submitted files for images.

## Rights and permissions

## About this article

### Cite this article

Hilgenfeld, B., Haueisen, J. Simultaneous suppression of disturbing fields and localization of magnetic markers by means of multipole expansion.
*BioMag Res Tech* **2**, 6 (2004). https://doi.org/10.1186/1477-044X-2-6

Received:

Accepted:

Published:

DOI: https://doi.org/10.1186/1477-044X-2-6